M400 Series Inverted Metallurgical Microscopy System

M400 Series Inverted Metallurgical Microscope is a
professional Metallurgical Testing Instrument for observation & analysis
the characteristics of metallurgy, Mineralogy, crystal and micro-electronics,
etc. It is widely used in Factories, Universities, Scientific research organization
and Electronic industry.

Main Features:

•
shape-designed base provides the high stability; Ergonomic Configuration
Design achieves simple and convenient operation

•  Both reflection and transmission illumination are
available; Bright field and Polarizing observation under illumination of
reflection light; bright field observation under  transmission light. Polarizing

Outfit is already equipped for application.

Standard
Specification -----M400 Series

Standard Specification

M400

M403

M410

M413

Viewing Head

Compensation
Binocular Head       Inclined
at45º Interpupillary Distance:55mm -75mm

√

OPTIONAL

√

OPTIONAL

Compensation
Trinocular Head     Inclined
45º   Interpupillary Distance:55mm -75mm

OPTIONAL

√

OPTIONAL

√

Eyepiece

WF10X/20mm

√

√

OPTIONAL

OPTIONAL

WF10X/22mm

OPTIONAL

OPTIONAL

√

√

WF10X/20mm
w/reticule 0.1mm

OPTIONAL

√

OPTIONAL

√

Nosepiece

Quadruple
nosepiece

√

√

√

√

Objective

Metallurgical Objectives    4xÁ 0.1W.D
.25mm;  10xÁ 0.25W.D .11mm  ,
20 xÁ 0.4W.D .9mm;
40x¡Á 0.6W.D.3.8mm

√

√

OPTIONAL

OPTIONAL

Infinity
metallurgical objectives
4 xÁ 0.1W.D .25mm ,  10 xÁ
0.25W.D .11mm ,  20xÁ
0.4W.D .9mm;   40 xÁ 0.6W.D.3.8mm

OPTIONAL

OPTIONAL

√

√

50 xÁ /0.75W.D. 1.9mm 80 xÁ
(S)/0.9W.D. 0.9mm

OPTIONAL

OPTIONAL

OPTIONAL

OPTIONAL

Stage

Double
layers mechanical stage; Stage size:242mmX 172mm Moving range :75mmX 50mm

√

√

√

√

Condenser

Abbe N. A.1.25
condenser with iris diaphragm &filter

√

√

√

√

Focusing

Coaxial
coarse& fine focusing adjustment with rack and pinion mechanism Fine
focusing scale : 0.002mm

√

√

√

√

Illumination

Epi-Kohler
illumination With aperture iris diaphragm and field iris diaphragm

√

√

√

√

Transmission
light, 12V/30W.AC85V-230V Adjustable brightness

√

√

√

√

Polarizing Outfit

Analyzer
rotatable 360 °,polarizer & Analyzer can be
slided in / out of the optical path

√

√

√

√

Filter

Blue
,green, yellow

√

√

√

√

Checking tool

0.01mm
micrometer

√

√

√

√

Optional

DCM3.0
digital eyepiece

OPTIONAL

OPTIONAL

OPTIONAL

OPTIONAL

Accessories

DCM130
Digital eyepiece

OPTIONAL

OPTIONAL

OPTIONAL

OPTIONAL

Photo
attachment

OPTIONAL

OPTIONAL

OPTIONAL

OPTIONAL

.

Components and Accessories

MET40000

Compensation Binocular Head  Inclined at45 º

MET40600

Collector

MET40002

Compensation Trinocular
Head ,Inclined at45 º

MET40701

Epi-Kohler
illumination With aperture iris diaphragm and field iris diaphragm

MET40101

WF10X/20mm

MET40702

Transmission light system, 12V/30W. AC85V-230V

MET40102

WF10X/22mm

MET40800

12v/30w halogen bulb

MET40103

WF10X/20mm w/reticule 0.1mm

MET40810

Blue ,green, yellow Filter kit

MET40200

Outward Quadruple nosepiece

MET40900

0.01mm micrometer

MET40301

Metallurgical Objectives

MET40302

I nfinity metallurgical
objective s

4 XA 0.1W.D .25mm;10 XÁ 0.25W.D .11mm

4XÁ 0.1W.D .25mm;
10 XÁ 0.25W.D .11mm

20 XÁ 0.4W.D .9mm;40 XÁ
0.6W.D.3.8mm

20 XÁ 0.4W.D .9mm;
4 0 XÁ 0.6W.D.3.8mm

MET40303

Metallurgical
Objectives

MET40304

I nfinity metallurgical
objective s

50x/0.75 WD1.9mm,80x/0.9WD0.9mm

50x/0.75WD1.9mm,

80x/0.9WD0.9mm

MET40400

Double layers mechanical stage;

MET40405

Abbe N.
A.1.25 condenser

Stage size:180mmX 150mm

with iris diaphragm &filter

MET40500

Coaxial Focusing Adjustment System

MET40801

DCM300digital eyepiece

MET40910

Photo attachment

MET40802

DCM130 Digital eyepiece